iMM-6 iDevice Calibrated Measurement Microphone
Precision condenser microphone for critical measurement
Notify me when back in stock
Dayton Audio's IMM-6 is also compatible with Android and Microsoft devices that use a 3.5 mm TRRS jack. A headphone/line-out is built in for monitoring the recorded signal, and a kickstand elevates and angles the measurement capsule for more direct measurements. Each IMM-6 is ruggedly built with high-quality components. A sturdy carrying case with foam insert is included.
Specifications: • Capsule type: Precision 6 mm electret condenser • Polar Response: True Omnidirectional • Frequency response: 18-20,000 Hz (Calibrated) • Impedance: 200 ohms • Sensitivity at 1 kHz into 1K ohm: 10 mV/Pa (-40dBV, re. 0dB = 1V/Pa) • Maximum SPL for 1% THD @ 1000 Hz: 127 dB • S/N ratio: 70 dB (A-weighted, varies depending on host device) • Connector: Gold-plated 3.5 mm TRRS
IMM-6 file downloads not available on mobile. Please use a different device.
Help for Android and iPhone users:
There are many applications in the market that provide the smart phone user with audio measurement software. We do not recommend any specific apps, however, in an attempt to assist the new user of an iMM-6 we have listed links and some helpful pointers below. One for the iDevice/iPhone platform and the other for Android devices.
AudioTools app (from Studio Six Digital):
Here is the procedure for loading the iMM-6 calibration file into AudioTools by Studio Six Digital on an iPhone:
1) You need a Wi-Fi network where you can view other computers on the network. You also need a laptop PC, or a desktop PC connected directly to the wireless router via either a cable or a wireless networking card. Your iPhone and desktop PC will need to be able to 'see' each other to transfer files. Your wireless router should not be blocking port 9900. (If you haven't set your wireless router specifically to do this, it shouldn't be blocking this port.)
2) Make sure your iPhone has Wi-Fi enabled, and is connected to the same wireless network as your PC.
3) On your PC, download the iMM-6 serial calibration file (found on the bottom of the iMM-6 aluminum container) for your specific microphone and save it where you can easily locate it. You will need the serial number that is printed on the back of your iMM-6 carrying case to access the unique calibration file for your iMM-6.
4) On the iPhone, within AudioTools, navigate to the Input Sources screen and choose Microphone Calibration. The Calibration Screen will be displayed.
5) From the Calibration Screen, choose the Cal File option. Tap the Files button to bring up the Files screen.
6) At the top of the Files screen, there is a field labeled "URL", with http:// and an IP address (sequence of four three-digit numbers), plus a port number following the colon. (example: http://192.168.0.9:9900 – yours will probably be different)
7) From your PC which is connected to the wireless network, open up your web browser and type the full URL from the "URL" field into the address bar of the web browser. (Following the example above, you would type http://192.168.0.9:9900 into the address bar). This is the address for direct file transmission to your iPhone.
8) You should see a page appear with the heading "Documents on Strider" or something similar. Your iPhone is acting as a server on your WiFi network, and is serving this page to your PC. Click on the "calibration" link, which will navigate you to the "calibration" directory on your iPhone.
9) In the gray box labeled "Upload", click on the "Choose File" button to select your file. Find the iMM-6 calibration file you downloaded from above in step (3) and double-click on it.
10) In the gray box labeled "Upload", you should now see your iMM-6 calibration file listed. Click on "Send" to send the file to your iPhone. When done, the page will refresh and show your iMM-6 calibration file in the "calibration" directory.
11) Close your web browser. You are finished with the part of the upload that uses your PC.
12) On your iPhone, tap "Done" to return to the Calibration Files screen, and then tap the “Reload” button (round arrow) to refresh the view. You should see the calibration file appear.
13) To load the calibration file, tap the calibration file name and then tap "Apply".
14) The calibration file is now loaded. Tap "Done" to return to the Calibration screen.
These instructions are based on the instructions given on the Studio Six Digital website .
AudioTool by Bofinit Corporation
AudioTool supports loading a calibration file for the Dayton Audio iMM-6 microphone. These files have names like "99-0101.txt" when downloaded from from above. For use in AudioTool, you will simply need to rename the file so that it has the .cal file type (e.g. "99-0101.cal") and save it in the AudioTool directory on your Android phone. Then, in AudioTool, select the "Load Cal" option from the Menu, and choose the Dayton file from the list. The calibration data will be loaded, power summed to the 1/3 octave bins used by AudioTool, and saved in your Preferences - there is no further need to load the file whenever you start AudioTool, unless you change the calibration method or alter the calibration values manually.
• For use with Apple iPad, iPhone, iPod Touch via 3.5mm TRRS headset jack
• Compatible with Android and Windows Phone devices with TRRS jack
• Headphone/Line Out pass-through jack for test signals or listening
• Kickstand feature elevates and angles device and microphone capsule
• True omnidirectional pattern with calibrated flat frequency response
• Rugged construction with high-quality components
|Unit of Measure||Each|
|Warranty - Parts||5 Years|
|Warranty - Labor||5 Years|
|Product Shipping Weight (lbs.)||1.0000|
Be The First To Review This Product!
Help other The Loudspeaker Kit users shop smarter by writing reviews for products you have purchased.